Triple Water High Speed Dental Air Rotor Handpiece With Classic Non
1. Push Button High Speed Handpiece M4/B2
2. Working Air Pressure:0.20-0.25MPa(0.22MPa prefrred)
3. Rotation Speed:350,000-400,000R.P.M(Standard Head);
4. Bur Applicable:Φ1.595-1.600mm"
5. Spray:Triple water spray
1. Price of handpiece,dental original panamax push button air
turbine high speed handpiece.
2. 2-hole standard push button handpiece.
3. It has unique appearance design and fluent outline.
4. Smooth as silk, it has excellent texture.
5. With people-friendly design, easy for operation.
6. With best quality cartridge, easier for cleaning.
7. With high-speed turbine rotation, it has better performances.
8. Chucking power of chucking needle is 20N-45N.
9. The high speed handpiece is noise-free and stable.
10. Applicable Bur: 1.6mm and should be conformed to ISO standard.
11. Idling speed of handpiece should be no less than 370,000
rotations under working air pressure of 25- 30 pressure atmosphere.
12. Special for operations such as clamping high-speed dentistry
machine needle to drill and grind teeth in department of
stomatology, it is more professional.
13. The high speed handpiece is only for people who have dentistry
Consists of handpiece head, handle, caudal thread, cartridge.
1. Air pressure 0.25~0.30mpa,must be filtrated.
2. Use high speed dental bur with shaft diameter
Φ1.59~1.60mm×21-23mm and roghtness less than 0.8um.
3. The amount of handpieces spray or water spray should be able to
4. Hard glass fiber,can be repeatedly sterilized in high
temperautre and pressure.
Only used for drilling, molar of stomatology clamping high-speed
HPA009 Handpiecewe Provide Two Kind Of Head For Choice
dental handpiece advantage:
1: Mini( Standard /Torgue Head.
2: The machine maintenance is simple ,can be more than stopwatch
used alternately to extend service life.
3: Hanpieces for single water spray ,drainage holes and air flow
adopts precision edm.
The atomization effect is better .
4: Classic non-slip pattern , feel more comfortable.
5: The collet type movement compared with opne movement
more can guarantee the movemnt up and down bearing concentricity.
1. Considering safety first, please be careful to use it with full
2. Be limited to dentists who held the dental treatment
3. Strictly forbid decomposition and reconstruction by
4. Please run point testing on outside of patients oral beforehand.
If feel anomalies, please stop using immediately and contact with
our after-sales center.
5. When appears abnormal during using, please stop it immediately
and contact with our after-sales center.
6. It will cause a high noise using a center-vibration or bended
bur and a handpiece of which bearing has been worn, A long time in
this state will be harm to people’s hearing. Forbid using a
vibration or bended bur, if appear high noise, please replace the
shaft core early
7. Do not use a bending , cracking, deformation and unqualified
bur. If using it, may cause a snap, or a flying off to injury
accident during operation process.
8. Please ensure the bur chuck be clean. If with contaminant inside
of it, may causes bur clamping weaken or shaft core vibration.
9. Please run the bur under the provision rotation-speed by bur
10. Do not crash, especially to avoid falling.
11. Please install a bur or a standard stick before using.
12. Aslo remember to install a bur or a standard stick when not in
13. Please unload the bur and handpiece when the running stopped in